Ensure that chucks of the handpiece are cleaned regularly
Avoid leverage and jamming
Use eyeshield and aspiration
Maximum admissible speed has to be observed (indicated on each BUSCH box)
Instruments should reach the operating speed before they are applied to the work piece
Sufficient cooling water spray (minimum 50 ml/min) is to be used for certain preparation application in dental practice
Instruments have to be disinfected and, if necessary, sterilized before use.
Disinfection and cleaning:
Put instruments into a bath with suitable disinfection and cleansing liquid.
Afterwards, clean the instruments manually or by ultrasonic bath.
After cleaning the instruments, dry immediately.
Following the disinfection and cleaning, the instruments can be sterilized in the autoclave, chemiclave, or hot-air sterilizer.
Before the sterilization, the instruments have to be put into a case or foil permeable to germs and suitable for sterilization.
After the sterilization, the instruments in the case or foil should be kept in a dry and dust free place.
Standard warranty covered by seller against any manufacturing defect. In such events, please report to us within 7 days from date of delivery at email@example.com.
Overall Customer Rating:
0 customer reviews
Sign In Or Register to post your reviews. Sign Up