The CardioCare 2000 is an economically priced interpretive 12 channel electrocardiogram (ECG/EKG) machine that does not compromise on performance. Suitable for use in private practice, the ER, or hospitals. The CardioCare 2000 is designed for maximum ease of use and convenience. Software upgradeable and networkable, the CardioCare 2000 is an excellent choice for your diagnostic needs.
Simultaneous 12 Channel Electrocardiograph (ECG/EKG)
The CardioCare 2000 acquires 10 full seconds of ECG data simultaneously from 10 leads to produce an accurate 12 channel printout with analysis.
ECG Interpretation Program
Over 80 kinds of interpretation results based on the advanced Minnesota Code are provided to aid with your diagnosis. The interpretation algorithms are expanded with regular software updates.
A4 Full-Size Paper
The CardioCare 2000 will produce full-size A4 printouts in 1, 3, 6, and 12 channel formats.
Full-size print-outs improve readability and make it easier for filing in patient charts. A convenient Copy button lets you generate ECG copies in any of the available formats.
Low Maintenance Cost
The CardioCare 2000 uses normal fax paper which costs about ¼ the cost of ECG paper. It will even generate its own grid.
1 Channel + 1 Rhythm
3 Channel + 1 Rhythm
1 Channel (60 Sec 1 Rhythm)
BMS-Plus is an ECG software program that enables you to interface CardioCare 2000 to Windows-based computers through LAN. The ECG data from CardioCare 2000 is saved in either Bionet’s proprietary file format (.ecg), JPEG or PDF. The JPEG file format ensures compatibility with most EMR packages. BMS-Plus displays real-time ECG data on-screen, instantly performs a full set of measurements, and offers diagnostic aid. In addition, BMS-Plus can eliminate the need for costly thermal paper by printing a variety of comprehensive reports on standard printer/copier paper.
The CardioCare 2000 software is user upgradeable and is constantly being improved.
CardioCare 2000 Main Unit, 10 Lead ECG Patient Cable, Alligator Clips, Electrodes, Medi-Graph Paper, Hospital Grade Power Cord, Operation Manual, Diagnostics Guide Book, Quick Reference Guide and 4-Year Warranty Card.
10 lead 12 channel simulataneous 10-second resting ECG
296 (w) x 305.5 (h) x 92.5 (d) mm
Approximately 2.98 kg
3, 6, 12 channel and 1 channel (60 sec)
5, 10, 20 auto (I~aVF: 10, V1~V6:5) mm/mV
12.5, 25, 50 mm/s
AC (50/60 Hz, -20dB or better)
Muscle (25~35Hz, -3dB or better)
Base Line Drift (0.1Hz, -3dB or better)
Low pass filter (off, 40Hz, 100Hz, 150Hz)
2 x 16 character LCD display
Sensitivity, speed, filter status, HR, printing form, rhythm lead
Signal quality, power source, battery
Alphanumeric and symbol available
ID, name, age, sex, height, weight
Heart rate, PR, QRS, QT/QTc, P-R-T axis
Thermal print head, thermal roll paper
-Width: A4: 210mm or 8.5"
-Length: A4: 300mm or 11"
Effective writing width: 204mm
- Vertical: 8 dots/mm
- Horizontal: 16 dots/mm
Internal noise: 20uV(p-p) max
Input circuit: Floating input
Isolated and defibrillation protected
Input impedance: >10MQ
Input range: > +/- 5mV
Common Mode Rejection: > +100dB
DC offset voltage: > +/- 300mV
Time constant: 3.2 sec
Patient leakage current:
Frequency response: 0.05~150Hz
Signal Quality Control
Disconnected lead detection
Power Supply: AC or built-in Battery (option)
Power requirement: 100 - 240 Vac, 50/60Hz, 1.0 - 0.5A
Power consumption: 60w max
1 hour of normal use (approximately 100 automatic ECG printouts)
PC connection with RS-232 interface and LAN
Class 1, Type BF
Operating humidity: 30~85%
Operating temperature: 10° ~40° C
Atmospheric pressure: 70 ~ 106 KPa
Power Cable - 1 ea
Patient Cable - 1 ea
Electrode - 10 ea
Chart Paper - 1 ea
Operation Manual - 1 ea
Rechargeable Battery (ni-mh), Cart, Hanger, Bag, PC s/w, Limb electrode, Chest electrode, RS-232C null cagle, LAN cable (Ethernet-RJ-45)
* Specifications can be changed without prior notification.
Standard warranty covered by the seller against any manufacturing defect. In such events, please report to us within 7 days from the date of delivery at email@example.com.
Overall Customer Rating:
0 customer reviews
Sign In Or Register to post your reviews. Sign Up